This is an old revision of the document!
Please use the Transfers Page for information on data transfer.
Useful scripts are available on ops1hb. list_correlation.py scans the master schedules pages for the last few years and looks for a correlation date on experiments involving our stations. It writes a list of correlated experiments to correlated.txt on ops1hb. If an experiment appears in this list e.g.
cat correlated.txt | grep -i r1758
then the data can be erased from the module.
Also on ops1hb is data_summary.py which reads the master schedule, finds upcoming scheduled experiments with summary files, finds the data size for each station and writes this information to the Google module spreadsheet AuScope IVS Module Allocation.
Please enter any significant events related to the AuScope VLBI project, including observations, configuration changes, hardware upgrades etc.
After the receiver upgrade at Katherine, methanol maser observations (using
methgain.sh on flexbuffke) indicated significant pointing offsets in both Az and El. After 24 hours of observing the data were combined into a MATLAB file and a new incremental pointing model was fitted to the offsets. The combined Pointing model values were then calculated and a new mdlpo.ctl file was created and tested on pcfske. The result has been a reduction in median pointing offset from 0.048 to 0.014 degrees.
The new pointing model is installed on pcfske as mdlpo.ctl with appropriate comments.
The humidity sensor on the MET4 isn't working and is going off for repair.
On pcfsyg as root:
pcfsyg:/etc/init.d# update-rc.d -f metclient.sh remove pcfsyg:/etc/init.d# update-rc.d -f metserver.sh remove pcfsyg:/etc/init.d# ./metserver.sh stop pcfsyg:/etc/init.d# ./metclient.sh stop
Also commented out the call to wx in /usr2/proc/station.prc
After installing the maser post refurb, and SSR installation for aircon compressor, the aircon was turned off at 0452 on Fri 3/6/16 and a large fan was turned on. 0430UT on 6/6/16 air con was turned on and large fan off and maser room closed.
The above has been done to see if the maser aircon at Ke is affecting the clock jump problem. Cheers, Brett
This is to let you know that the VLBI antenna on the 13/14th December had the following work done 1. Az bullgear old grease removed - new grease applied. 2. El jack screw cover was removed and more new grease applied to the jack screw arm. The jack screw cover itself was cleaned but is in need of replacing very soon. It would be good to get this done when Brett is over next. Also I need to check the gear box oil and or maybe change the oil with him? 3. The azimuth grease nipples inside the pedestal and the grease nipples on the antenna itself have had grease pumped in. 4. A general clean up was done inside the pedestal and the air conditioning filters removed and cleaned.