skip to content
AuScope VLBI Operations Wiki
You are here:
Documentation for operators
This wiki is not maintained! Do not use this when setting up AuScope experiments!
Pointing and fringe checks were not performed, otherwise started right in time.
erc Status Monitor window shows wrong time: 1969.264.23.0.00 while it is really (according to formatter, FS, computer) is 2014.233.08.20.00. Execution of commands happens according to real time.
At around 16:45, cryogenic show signs of warm-up: T(70K phase)=84K, T(20K)=65K, He pressure goes down recently, but currently 305 psi
Frequent error: “Error being ignored: error qk -211 tsys value for device ic overflowed or was less than zero.” and “Error being ignored: error ch -308 vd total power integrator overflow”
recording to USN-0109
Peaks autocorrelation bandpasses in ch 14, 15 – RFI
recording to NRCAN-05
Persistent problem: WARNING: error an -103 pointing computer tracking errors are too large. ; WARNING: error qo -301 warning: onsource status is slewing!
recording to NRCAN-07
Internet connection is flaky, ERC loses connection every now and then, so often that even filling the checklist is problematic
Descending slope in the autocorrelation bandpass in channel 15 – RFI
recording to USN-0205
You could leave a comment if you were logged in.
· Last modified: 2014/08/21 21:40 by
Back to top