This wiki is not maintained! Do not use this when setting up AuScope experiments!
Disk VSN: | Data volume at beginning: 0.641 GB
1730UT Experiment started late. Bad Mark5B module. Originally thought it was the Mark5B itself going bad but got Peter to replace module and it worked (Lucas).
First scan 0403-132 at 17:33 UT
pcfsyg disk got 100% full. Had to delete monica points from 2011,2012,2013 and 2014. Freed up 30G of space. Need more permanent solution (Lucas)
03:50 recorded data 104 GB less than scheduled (warren)
13:00 UT - pcfsyg disk 100% full. Deleted some lines from the Monica log file (oMonica.log) eventually to free up some disk space. I don't think this will solve the problem completely though, as the log file appears to be growing very quickly. (Patrick)
22:30 UT Patrick discovered oMonica.log out of control writing 20 GB per hour due to Monica error connecting to weather on localhost:50001 A restart of Monica server on pcfsyg fixed this.(Warren and Patrick)
/home/www/auscope/opswiki/data/pages/handover/r1912.txt · Last modified: 2019/09/18 00:08 by Warren Hankey