Power failure at MtP. Dish came back on source at 2217UT.
70K stage temperature values shoot up from time to time. Display/sensor issue?
Power failure at around 2230UT. Restarted system at 2243UT. Scans 015-2229 and 015-2242 affected.
2250UT - Wind stowed is what the live page indicated. Back on at 2259UT. Scan 015-2250 missed.
0733UT mk5cn: time-out, connection closed occasionally happened after a few “recording stopped”. Recording started normally for the subsequent scans.
1606UT - looks like it fails to connect to mk5, auto correlation plot shows correct timestamp but scan number “-alpha-”, warning “mk5cn: error opening, mark 5 probably not running, see above for error” without error. disk_pos doesn't update. One of the Mk5hb light flickered, ip switching mk5hb made one of the light on module flickers. (Lim)
Problem not solved till the end of schedule. Scans very likely affected. (Lim)
/home/www/auscope/opswiki/data/pages/handover/rd2001.txt · Last modified: 2020/01/16 18:24 by Pradyumna Kiran Kummamuru